You are here
Regulation of Software as a Medical Device
Software is becoming increasingly important in medical devices. In addition, it is becoming more and more important as a medical device in its own right.
Medical devices are regulated in Australia by the Therapeutic Goods Administration (TGA). This includes software and mobile ‘apps’ that meet the definition of a medical device.
The following is intended to provide guidance on the regulation that applies to software and apps that meet the legislated definition of a medical device in Australia.
The definition of Software as a Medical Device
A software product is considered to be a medical device if it fits the definition of a medical device in section 41BD of the Therapeutic Goods Act 1989.
A medical device is:
- any instrument, apparatus, appliance, material or other article (whether used alone or in combination, and including the software necessary for its proper application) intended, by the person under whose name it is or is to be supplied, to be used for human beings for the purpose of one or more of the following:
- diagnosis, prevention, monitoring, treatment or alleviation of disease;
- diagnosis, monitoring, treatment, alleviation of or compensation for an injury or disability;
- investigation, replacement or modification of the anatomy or of a physiological process;
- control of conception;
and that does not achieve its principal intended action in or on the human body by pharmacological, immunological or metabolic means, but that may be assisted in its function by such means.
Many types of software meet this definition and are therefore regulated. Some examples of SaMD include:
- smart phone apps that calculate insulin doses based on a patient's blood glucose levels
- X-ray image-processing software
- software that uses information about a patient to make a diagnosis.
SaMD may be used with or in different computing platforms such as:
- Mobile phones*
*Note that for the SaMD examples the mobile phone, computer or tablet is not itself a medical device.
Health software apps that are not a medical device
Many mobile apps are simply sources of information, or tools to manage a healthy lifestyle. The TGA does not regulate health and lifestyle apps and software that do not meet the definition of a medical device.
Apps that use accessories connected to, or embedded in, the platform
Some mobile apps require the use of physical accessories that are connected to, or embedded in, the computing platform. These can be sensors that plug into a port on the platform, or features that are provided in the platform such as speakers or a camera. This results in the combination of software, accessory and the computing platform becoming a medical device.
One example is a glucose meter that reads blood test strips and plugs into a smartphone to display and store the results. In this case, the combination of meter, smartphone and app, is a medical device. The app is regulated as part of the glucose meter device.
Software that is part of a device
Medical device software that is an integral part of a physical device (for example embedded software or firmware in a cardiac pacemaker) is considered to be part of that device and is not regulated separately.
Apps that control a medical device
Some mobile apps can control or adjust a medical device through Bluetooth or WiFi features. These apps are considered to be SaMD because they are accessories to the medical device. They are regulated at the same risk classification level as the medical device they control.
Software as a Medical Device (SaMD) regulation in Australia
The regulation of medical devices is risk-based. This means that the level of scrutiny and oversight by the TGA will vary according to the level of risk that the product represents to the patient or healthcare professional using it.
The current regulations do not adequately capture all SaMD under the rules for this risk classification. The potential risks arising from SaMD products can be low, medium or high depending on the intended purpose of the SaMD; however, at the present time many SaMD products are classified as low risk even though the potential risk for users may be higher. The TGA will soon be consulting on changes to the regulations to address this issue.
SaMD products must be included on the Australian Register of Therapeutic Goods (ARTG) before they are supplied in Australia.
The therapeutic goods legislation requires manufacturers of SaMD products (unless they are Class 1 - the lowest risk classification) to obtain Conformity Assessment certification to allow inclusion in the ARTG. All medical devices, irrespective of classification, are required to meet the Essential Principles for safety and performance.
International regulatory activities
The TGA is a founding member of the International Medical Device Regulators Forum (IMDRF), a group of medical device regulators from around the world who meet regularly to address current challenges in regulating medical devices, and to accelerate harmonisation of regulation.
In 2013, the IMDRF established working group dedicated to Software as Medical Device, and they have been working on the challenges associated with it. As an active member of that working group, the TGA contributed to the four published technical documents relating to the regulation of SaMD:
- Software as a Medical Device (SaMD): Key Definitions
- Software as a Medical Device: Possible Framework for Risk Categorization and Corresponding Considerations
- Software as a Medical Device (SaMD): Application of Quality Management System
- Software as a Medical Device (SaMD): Clinical Evaluation
These documents were consulted internationally before being finalised and are available on the IMDRF Documents webpage (Technical documents section). Together, they cover the definition of Software as a Medical Device, as well as how existing regulatory tools such as quality management and clinical evidence can be applied to it.
The TGA recognises that the existing regulatory framework for medical devices in Australia may not be well structured to address the potential public health risks posed by SaMD products. Therefore, the TGA is currently reviewing the regulatory framework for SaMD in Australia. Part of that is seeing where we can harmonise with the published IMDRF technical documents for SaMD.
The TGA has people dedicated to SaMD and other challenges associated with regulating digital medical devices. If you have questions about the regulation of SaMD or other digital medical technologies, please contact: firstname.lastname@example.org
More information on SaMD will be coming soon.